<!doctype HTML public "-//W3C//DTD HTML 4.0 Frameset//EN">

<html>

<head>
<title>SENSe:CORRection:COLLect</title>
<meta http-equiv="content-type" content="text/html; charset=windows-1252">
<meta name="generator" content="RoboHelp by eHelp Corporation www.ehelp.com">
<link rel="stylesheet" href="../../Romes_ns.css"><script type="text/javascript" language="JavaScript" title="WebHelpSplitCss">
<!--
if (navigator.appName !="Netscape")
{   document.write("<link rel='stylesheet' href='../../Romes.css'>");}
//-->
</script>
<style type="text/css">
<!--
p.whs1 { x-condition:OLH; }
img_whs2 { border:none; width:42px; height:18px; float:none; x-condition:OLH; border-style:none; }
hr.whs3 { float:aligncenter; width:50%; x-condition:OLH; }
img_whs4 { border:none; width:47px; height:16px; float:none; border-style:none; }
img_whs5 { border:none; width:36px; height:37px; float:none; border-style:none; }
td.whs6 { background-color:#cbcbcb; }
td.whs7 { border-left-width:4px; border-top-width:4px; border-right-width:4px; border-bottom-width:4px; background-color:#cbcbcb; }
td.whs8 { background-color:#cbcbcb; border-left-width:4px; border-left-style:none; border-top-width:4px; border-top-style:none; border-right-width:4px; border-right-style:none; border-bottom-width:4px; border-bottom-style:none; }
td.whs9 { background-color:#cbcbcb; border-left-width:4px; border-top-width:4px; border-right-width:4px; border-bottom-width:4px; }
td.whs10 { border-left-style:none; border-top-style:none; border-right-style:none; border-bottom-style:none; background-color:#cbcbcb; border-left-width:4px; border-top-width:4px; border-right-width:4px; border-bottom-width:4px; }
p.whs11 { font-family:Arial, sans-serif; }
h5.whs12 { x-condition:OLH; }
-->
</style><script type="text/javascript" language="JavaScript" title="WebHelpInlineScript">
<!--
function reDo() {
  if (innerWidth != origWidth || innerHeight != origHeight)
     location.reload();
}
if ((parseInt(navigator.appVersion) == 4) && (navigator.appName == "Netscape")) {
	origWidth = innerWidth;
	origHeight = innerHeight;
	onresize = reDo;
}
onerror = null; 
//-->
</script>
<style type="text/css">
<!--
div.WebHelpPopupMenu { position:absolute; left:0px; top:0px; z-index:4; visibility:hidden; }
-->
</style><script type="text/javascript" language="javascript1.2" src="../../whmsg.js"></script>
<script type="text/javascript" language="javascript" src="../../whver.js"></script>
<script type="text/javascript" language="javascript1.2" src="../../whproxy.js"></script>
<script type="text/javascript" language="javascript1.2" src="../../whutils.js"></script>
<script type="text/javascript" language="javascript1.2" src="../../whtopic.js"></script>
</head>
<body><script type="text/javascript" language="javascript1.2">
<!--
if (window.gbWhTopic)
{
	if (window.addTocInfo)
	{
	addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nAQUire");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nAQUire:SELected");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nACQuire:SAVE");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nACQuire:SAVE:DEFault");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nAUTO");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nAUTO:CKIT");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nAUTO:PORTs");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nAUTO:PORTs:CONNection");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nCONNection");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nCONNection:PORTs");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nDELete");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nMETHod");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nMETHod:DEFine");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nSAVE");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nSAVE:DEFault");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nSAVE:SELected");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nSAVE:SELected:DEFault");
addTocInfo("Command Reference\nSCPI Command Reference\nSENSe\n[SENSe<Ch>:]CORRection\n[SENSe:]CORRection:COLLect\nSCONnection");
addButton("show",BTN_TEXT,"Show","","","","",0,0,"","","");

	}
	if (window.writeBtnStyle)
		writeBtnStyle();

	if (window.writeIntopicBar)
		writeIntopicBar(1);

	if (window.setRelStartPage)
	{
	setRelStartPage("../../Start.htm");

		autoSync(1);
		sendSyncInfo();
		sendAveInfoOut();
	}
}
else
	if (window.gbIE4)
		document.location.reload();
//-->
</script><script type="text/javascript" language="JavaScript1.2" src="../../ehlpdhtm.js" x-save-method="compute-relative" name="EHELP_DHTM"></script>
<script type="text/javascript"
		language=JavaScript1.2>
<!-- 
if( typeof( kadovInitEffects ) != 'function' ) kadovInitEffects = new Function();if( typeof( kadovInitTrigger ) != 'function' ) kadovInitTrigger = new Function();if( typeof( kadovFilePopupInit ) != 'function' ) kadovFilePopupInit = new Function();if( typeof( kadovTextPopupInit ) != 'function' ) kadovTextPopupInit = new Function();
 //-->
</script>


<form>


<h5>[SENSe&lt;Ch&gt;:]CORRection:COLLect... &nbsp;&nbsp;
<a href="javascript:void(0);" onClick="var e; if (parseInt(navigator.appVersion)>=4){e=event;}  if (parseInt(navigator.appVersion) >=2 && typeof(PopupMenu_Invoke) == 'function') PopupMenu_Invoke(e,'','Calibration','../../MMI_Reference/Channel/Calibration/Channel_-_Calibration.htm','Calibration (program example)','../../Program_Examples/Condensed_Program_Examples/Calibration.htm','Calibration Kits','../../System_Overview/Calibration/Calibration_Kits.htm','SENSe:CORRection','SENSe_CORRection.htm');return false;" title="Related Topics"><img src="../../ss_btn_related_topics.gif" border=0 height=20 width=96 alt="Related Topics"></a>


 </h5>

<p>This subsystem controls the system error correction and manages calibration 
 kits. &nbsp;</p>

<p class=space>&nbsp;&nbsp;&nbsp;</p>

<p class="whs1">Command tree: <a class="dropspot" href="javascript:kadovTextPopup(this)" id="a15" style="x-condition: OLH; position: relative;">[SENSe&lt;Ch&gt;:]CORRection:COLLect</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a15');
//--></script></p>

<div class="droptext" id="POPUP170545336" style="display: 'none'; " x-ss-type="dropdown">
 <p class=Befehl
	style="margin-left: 144pt;"><a href="SENSe_CORRection_COLLect.htm#AQUire"
									id=a1
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[:ACQuire]</a></p>
 <p class=Befehl
	style="margin-left: 162pt;"><a href="SENSe_CORRection_COLLect.htm#AQUire_SAVE"
									id=a40
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:RSAVe</a></p>
 <p class=Befehl
	style="margin-left: 180pt;"><a href="#AQUire_SAVE_DEFault"
									id=a5
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:DEFault</a></p>
 <p class=Befehl
	style="margin-left: 162pt;"><a href="SENSe_CORRection_COLLect.htm#AQUire_SELected"
									id=a3
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:SELected</a></p>
 <p class=Befehl
	style="margin-left: 144pt;"><a href="SENSe_CORRection_COLLect.htm#AUTO"
									id=a35
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:AUTO</a></p>
 <p class=Befehl
	style="margin-left: 162pt;"><a href="SENSe_CORRection_COLLect.htm#AUTO_CKIT"
									id=a58
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:CKIT</a></p>
 <p class=Befehl
	style="margin-left: 162pt;"><a href="#AUTO_PORTs"
									id=a2
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:PORTs</a></p>
 <p class=Befehl
	style="margin-left: 180pt;"><a href="SENSe_CORRection_COLLect.htm#AUTO_PORTs_CONNection"
									id=a48
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:CONNection</a></p>
 <p class=Befehl
	style="margin-left: 144pt;"><a href="SENSe_CORRection_COLLect.htm#CONNection"
									id=a4
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:CONNection</a></p>
 <p class=Befehl
	style="margin-left: 162pt;"><a href="SENSe_CORRection_COLLect.htm#CONNection_PORTs"
									id=a11
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:PORTs</a></p>
 <p class=Befehl
	style="margin-left: 144pt;"><a href="SENSe_CORRection_COLLect.htm#DELete"
									id=a13
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:DELete</a></p>
 <p class=Befehl
	style="margin-left: 144pt;"><a href="SENSe_CORRection_COLLect.htm#METHod"
									id=a19
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:METHod</a></p>
 <p class=Befehl
	style="margin-left: 162pt;"><a href="SENSe_CORRection_COLLect.htm#METHod_DEFine"
									id=a20
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:DEFine</a></p>
 <p class=Befehl
	style="margin-left: 144pt;"><a href="SENSe_CORRection_COLLect.htm#SAVE"
									id=a21
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:SAVE</a></p>
 <p class=Befehl
	style="margin-left: 162pt;"><a href="SENSe_CORRection_COLLect.htm#SAVE_DEFault"
									id=a22
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:DEFault</a></p>
 <p class=Befehl
	style="margin-left: 162pt;"><a href="SENSe_CORRection_COLLect.htm#SAVE_SELected"
									id=a23
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:SELected</a></p>
 <p class=Befehl
	style="margin-left: 180pt;"><a href="SENSe_CORRection_COLLect.htm#SAVE_SELected_DEFault"
									id=a24
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:DEFault</a></p>
 <p class=Befehl
	style="margin-left: 144pt;"><a href="#SCONnection"
									id=a6
									style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">:SCONnection</a></p>
</div>
<p class=Linkline
	style="margin-bottom: 0pt; x-condition: OLH;"><img src="../../small_images/helpwoval3.gif" x-maintain-ratio="TRUE" width="42px" height="18px" border="0" class="img_whs2"> <a href="../../Program_Examples/Condensed_Program_Examples/Calibration.htm" id="a42" style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">Program 
 example: Calibration</a><script type="text/javascript"
													language=JavaScript1.2><!--
kadovInitEffects('a42');
//--></script></p>


<hr align="center" width="50%" class="whs3">


<h6><a name=AQUire></a><img src="../../small_images/ZVR.gif" x-maintain-ratio="TRUE" width="47px" height="16px" border="0" class="img_whs4"> [SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect[:ACQuire]</span><br>
THRough | OPEN1 | OPEN2 | OPEN12 | SHORT1 | SHORT2 | SHORT12 | MATCH1 | 
 MATCH2 | MATCH12 | NET | ATT | IMATCH12 | REFL1 | REFL2 | SLIDe1 | SLIDe2 
 | SLIDe12 | LINE1 | LINE2 | M1O2 | O1M2 | M1S2 | S1M2 | OSHort1 | OSHort2 
 | UTHRough[, &lt;Dispersion&gt;, AUTO | &lt;delay | phase&gt;] </h6>

<p>Starts a calibration measurement in order to acquire measurement data 
 for the selected standards. The standards are reflection or transmission 
 standards and must be connected to port 1 or 2 of the analyzer.</p>

<p><img src="../../small_images/tip.jpg" x-maintain-ratio="TRUE" width="36px" height="37px" border="0" class="img_whs5"> Use the generalized command <span style="font-family: 'Courier New', monospace;"><a 
 href="#AQUire_SELected"
	id=a38
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect[:ACQuire]:SELected</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a38');
//--></script> 
 to obtain measurement data at arbitrary analyzer ports. </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. If unspecified 
 the numeric suffix is set to 1.</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>THRough ... UTHRough</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Standard types: <span style="font-style: italic;"><I>Through, 
 Open, Short, Match (MATCH12 and IMATCH12 are synonymous), Symmetric Network 
 (NET), Attenuation (ATT), Reflect, Sliding Match (SLIDe), Line, Match/Open 
 (M1O2, O1M2), Match/Short (M1S2, S1M2), Open/Short (OSHort), Unknown Through 
 (UTHRough)</I></span>. &nbsp;</p>
<p class=Tabtext>The numbers in the parameters denote the analyzer ports. 
 Two numbers 12 mean that two separate calibrations are performed at ports 
 1 and 2. &nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>ON</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;Dispersion&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Optional status parameter for UTHRough standard: </p>
<p class=Tabtext>OFF &#8211; 
 Unknown through standard is non-dispersive <br>
ON &#8211; 
 Unknown through standard is non-dispersive </td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>OFF</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>AUTO | &lt;delay | phase&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Optional entry of delay time or phase for UTHRough standard: 
 &nbsp;</p>
<p class=Tabtext>AUTO &#8211; 
 The analyzer determines the delay time or phase during the calibration 
 sweep<br>
&lt;delay or phase&gt; &#8211; 
 entry of the delay time in ps (for non-dispersive standards) or of an 
 estimate of the phase at the start frequency of the sweep in deg (for 
 dispersive standards). If an estimate of the start phase is entered, the 
 analyzer uses the <span style="font-weight: bold;"><B>calculated</B></span> value 
 which is closest to the estimate. &nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>AUTO</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Confirmed (with device-specific standards), no query</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs6">
<p class=Tabtext><span style="font-family: Arial, sans-serif;">See</span><span 
 style="font-family: 'Courier New', monospace;"> </span><a href="#SAVE"
															id=a29
															style="color: #0000ff;
	font-family: 'Courier New', monospace;
	/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]<span 
 class=dropspot
	style="color: #0000ff; font-family: 'Courier New', monospace;"><font color=blue >CORRection:COLLect:SAVE</font></span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a29');
//--></script><span 
 style="font-family: 'Courier New', monospace;">. </span></p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=AQUire_SAVE></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect[:ACQuire]:RSAVe 
 &lt;Boolean&gt;</span></h6>

<p>Activates or deactivates the calibration mode where the raw measurement 
 data of the standards is stored after the calibration is completed. The 
 setting is valid for the current calibration, where it overwrites the 
 global setting (<span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#AQUire_SAVE"
	id=A39
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect[:ACQuire]:RSAVe:DEFault</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A39');
//--></script>). 
 A new calibration deletes the calibration data acquired in previous calibrations. 
 </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;Boolean&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>ON | OFF - Keep measurement data on or off. &nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>OFF </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, command or query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs7">
<p class=Example
	style="color: #000000;"><span style="color: #000000;"><font color=black >CORR:COLL:RSAV:DEF 
 ON</font></span></p>
<p class=Exampletext>Generally keep the raw measurement data of the standards 
 after the calibration is completed. &nbsp;</p>
<p class=Example>CORR:COLL:METH:DEF 'Test',RSHort,1</p>
<p class=Exampletext>Select a one-port normalization at port 1 with a short 
 standard as calibration type.</p>
<p class=Example>CORR:COLL:SEL SHOR,1</p>
<p class=Exampletext>Measure a short standard connected to port 1 and store 
 the raw measurement results of this standard. </p>
<p class=Example
	style="color: #000000;"><span style="color: #000000;"><font color=black >CORR:COLL:RSAV 
 OFF</font></span></p>
<p class=Exampletext>To save disk space, delete the current raw calibration 
 data after the calibration is completed. </p>
<p class=Example>CORR:COLL:SAVE:SEL </p>
<p class=Exampletext>Calculate the system error correction data and apply 
 them to the active channel.</p>
<p class=space>&nbsp;&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=AQUire_SAVE_DEFault></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect[:ACQuire]:RSAVe:DEFault 
 &lt;Boolean&gt;</span></h6>

<p>Activates or deactivates the calibration mode where the raw measurement 
 data of the standards is stored after the calibration is completed. The 
 setting remains valid for all subsequent calibrations until it is changed 
 explicitly. A new calibration deletes the calibration data acquired in 
 previous calibrations. </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;Boolean&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>ON | OFF - Keep measurement data on or off. &nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>OFF </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, command or query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs7">
<p class=Example
	style="font-family: 'Courier New', monospace;"><span style="font-family: Verdana, sans-serif;">See</span> 
 <a href="#AQUire_SAVE"
	id=a37
	style="color: #0000ff; /*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]<span 
 class=dropspot
	style="color: #0000ff;"><font color=blue >CORRection:COLLect[:AQUire]:RSAVe</font></span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a37');
//--></script>. 
 </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=AQUire_SELected></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect[:ACQuire]:SELected</span><br>
THRough | OPEN | SHORt | MATCh | NET | ATT | REFL | SLIDE | LINE | UTHRough,&lt;port_no&gt;,&lt;port_no&gt;[, 
 &lt;Dispersion&gt;, AUTO | &lt;delay | phase&gt;]</h6>

<p>Starts a calibration measurement in order to acquire measurement data 
 for the selected standards. The standards are reflection or transmission 
 standards and can be connected to arbitrary analyzer ports. </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>THRough ... UTHRough</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Standard types: <span style="font-style: italic;"><I>Through, 
 Open, Short, Match, Symmetric Network (NET), Attenuation (ATT), Reflect, 
 Sliding Match (SLIDe), Line, Unknown Through</I></span>. &nbsp;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;port_no&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Port numbers of the analyzer. For a transmission standard 
 (through, line, attenuation, symmetric network) the input and output port 
 numbers must be specified, for reflection standards, only one port number 
 is required. </td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;Dispersion&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Optional status parameter for UTHRough standard: </p>
<p class=Tabtext>OFF &#8211; 
 Unknown through standard is non-dispersive <br>
ON &#8211; 
 Unknown through standard is non-dispersive </td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>OFF</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>AUTO | &lt;delay | phase&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Optional entry of delay time or phase for UTHRough standard: 
 &nbsp;</p>
<p class=Tabtext>AUTO &#8211; 
 The analyzer determines the delay time or phase during the calibration 
 sweep<br>
&lt;delay or phase&gt; &#8211; 
 entry of the delay time in ps (for non-dispersive standards) or of the 
 phase at the start frequency of the sweep in deg (for dispersive standards) 
 </td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>AUTO</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, no query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs7">
<p class=Example
	style="font-family: 'Courier New', monospace;">See <a href="#SAVE_SELected"
															id=a9
															style="color: #0000ff; /*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]<span 
 class=dropspot
	style="color: #0000ff;"><font color=blue >CORRection:COLLect:SAVE:SELected</font></span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a9');
//--></script>. 
 </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=AUTO></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:AUTO</span><br>
'&lt;file_name&gt;',&lt;port_no&gt;{,&lt;port_no&gt;}</h6>

<p>Selects and initiates a one-port, two-port, three-port or four-port 
 <a HREF="../../System_Overview/Calibration/Automatic_Calibration.htm"
	id=A36
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">automatic 
 calibration</a><script type="text/javascript"
											language=JavaScript1.2><!--
kadovInitEffects('A36');
//--></script><span 
 style="color: #ff0000;"> <font color=red ></font></span>at arbitrary analyzer and cal unit ports. 
 </p>

<p><img src="../../small_images/tip.jpg" x-maintain-ratio="TRUE" width="36px" height="37px" border="0" class="img_whs5">If the test setup contains a high attenuation the analyzer 
 may fail to detect the cal unit ports connected to each of its ports. 
 In this case use the extended command <span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#AUTO_PORTs"
	id=A53
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:AUTO:PORTs</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A53');
//--></script><span 
 class=dropspot>.</span> </p>

<p>If several calibration units are connected, use <span style="font-family: 'Courier New', monospace;"><a 
 HREF="../SYSTem/SYSTem.htm#RDEVice_AKAL_ADDRess"
	id=A54
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">SYSTem:COMMunicate:RDEVice:AKAL:ADDRess</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A54');
//--></script></span> 
 to select a unit for the calibration. </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>'&lt;file_name&gt;'</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Name and (possibly) directory of the cal kit file to be 
 used for the automatic calibration (string parameter): </p>
<ul>
	
	<li class=kadov-p-Clist><p class=list>If nothing is specified (empty 
 string ' '), the default cal kit file stored in the calibration unit is 
 used. This file is also used in manual control.</p></li>
	
	<li class=kadov-p-Clist><p class=list>A NWA cal kit file name *.calkit 
 without path denotes a specific cal kit file stored in the Calibration 
 Unit. </p></li>
	
	<li class=kadov-p-Clist><p class=list>A NWA cal kit file name *.calkit 
 with path denotes a specific cal kit file stored in an arbitrary directory. 
 </p></li>
</ul>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;port_no&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Port numbers of the analyzer. For an n-port automatic 
 calibration (n = 1 to 4), n arbitrary (not necessarily consecutive) port 
 numbers must be specified. The analyzer automatically detects the calibration 
 unit ports connected to each analyzer port. &nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, no query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs7">
<p class=Example><span class=dropspot>CORR:COLL:AUTO '', 1, 2, 4 </span></p>
<p class=Exampletext>Perform an automatic 3-port TOSM calibration at test 
 ports 1, 2, and 4 using the calibration unit's default calibration kit 
 file and arbitrary test ports of the cal unit. </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=AUTO_CKIT></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:AUTO:CKIT</span><br>
'&lt;file_name&gt;'</h6>

<p>Generate a cal kit file with the specified name containing the cal kit 
 data of the active calibration unit (<span style="font-family: 'Courier New', monospace;"><a 
 href="../SYSTem/SYSTem.htm#RDEVice_AKAL_ADDRess"
	id=a55
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">SYSTem:COMMunicate:RDEVice:AKAL:ADDRess</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a55');
//--></script></span>). 
 The cal kit file can be stored in the calibration unit or written to a 
 directory on the analyzer. </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel (not relevant 
 for this command). </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>'&lt;file_name&gt;'</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Name and (possibly) directory of the created cal kit file 
 (string parameter, extension *.calkit): </p>
<ul>
	
	<li class=kadov-p-Clist><p class=list>If no path and directory is specified, 
 the file is stored in the calibration unit (see also <span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#AUTO"
	id=A56
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:AUTO</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A56');
//--></script>). 
 The factory calibration data on the unit is not overwritten. </p></li>
	
	<li class=kadov-p-Clist><p class=list>If a path and directory is specified, 
 the file is stored in the directory on the analyzer's hard disk. </p></li>
</ul>
<p>The default directory (<span style="font-family: 'Courier New', monospace;"><a 
 HREF="../MMEMory/MMEMory.htm#CDIRecotry"
	id=A57
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">MMEMory:CDIRectory</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A57');
//--></script></span>) 
 is not used. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, no query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs7">
<p class=Example><span class=dropspot>CORR:COLL:AUTO:CKIT 'AutoCalChar.calkit' 
 </span></p>
<p class=Exampletext>Generate a cal kit file AutoCalChar.calkit for the 
 active calibration unit and store it internally. </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=AUTO_PORTs></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:AUTO:PORTs</span><br>
'&lt;file_name&gt;',&lt;analyzer_port_no&gt;,&lt;cal_unit_port_no&gt;{,&lt;analyzer_port_no&gt;,&lt;cal_unit_port_no&gt;}</h6>

<p>Selects and initiates a one-port, two-port, three-port or four-port 
 <a href="../../System_Overview/Calibration/Automatic_Calibration.htm"
	id=a41
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">automatic 
 calibration</a><script type="text/javascript"
											language=JavaScript1.2><!--
kadovInitEffects('a41');
//--></script><span 
 style="color: #ff0000;"> <font color=red ></font></span>at arbitrary analyzer and calibration 
 unit ports. </p>

<p><img src="../../small_images/tip.jpg" x-maintain-ratio="TRUE" width="36px" height="37px" border="0" class="img_whs5">This command is necessary if the analyzer fails to detect 
 the cal unit ports connected to each of its ports (e.g. because of a high 
 attenuation in the test setup). If auto-detection works you can use the 
 simpler command <span style="font-family: 'Courier New', monospace;"><a 
 href="#AUTO"
	id=a44
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:AUTO</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a44');
//--></script>. 
 </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>'&lt;file_name&gt;'</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Name and (possibly) directory of the cal kit file to be 
 used for the automatic calibration (string parameter): </p>
<ul>
	
	<li class=kadov-p-Clist><p class=list>If nothing is specified (empty 
 string ' '), the default cal kit file stored in the calibration unit is 
 used. This file should be used to operate the calibration unit like in 
 manual control; specifying another cal kit file is primarily for service 
 purposes. </p></li>
	
	<li class=kadov-p-Clist><p class=list>A NWA cal kit file name *.calkit 
 without path denotes a specific cal kit file stored in the Calibration 
 Unit. </p></li>
	
	<li class=kadov-p-Clist><p class=list>A NWA cal kit file name *.calkit 
 with path denotes a specific cal kit file stored in an arbitrary directory. 
 </p></li>
</ul>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;analyzer_port_no&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Port numbers of the analyzer. For an n-port automatic 
 calibration (n = 1 to 4), n arbitrary (not necessarily consecutive) port 
 numbers must be specified. </td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;cal_unit__port_no&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Port numbers of the cal unit. For an n-port automatic 
 calibration (n = 1 to 4), n arbitrary (not necessarily consecutive) port 
 numbers must be specified. It is possible to combine arbitrary (not necessarily 
 matching) pairs of analyzer and cal unit ports. &nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, no query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs7">
<p class=Example><span class=dropspot>CORR:COLL:AUTO:PORT '', 1, 2, 2, 
 4, 4, 1 </span></p>
<p class=Exampletext>Perform an automatic 3-port TOSM calibration at the 
 analyzer test ports 1, 2, and 4 using the calibration unit's default calibration 
 kit file and ports 2, 4, and 1 of the cal unit. </p>
<p class=space>&nbsp;</td></tr>
</table>


<hr align="center" width="50%" class="whs3">


<h6><a name=AUTO_PORTs_CONNection></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:AUTO:PORTs:CONNection?</span></h6>

<p>Returns the assignment between the network analyzer ports and the ports 
 of the connected <a href="../../System_Overview/Calibration/Automatic_Calibration.htm"
					id=a47
					style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">automatic 
 calibration</a><script type="text/javascript"
											language=JavaScript1.2><!--
kadovInitEffects('a47');
//--></script> 
 unit. </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel (not relevant 
 for this command). </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, query only</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs7">
<p class=Example><span class=dropspot>CORR:COLL:AUTO:PORT '', 1, 2, 2, 
 4, 4, 1 </span></p>
<p class=Exampletext>Perform an automatic 3-port TOSM calibration at the 
 analyzer test ports 1, 2, and 4 using the calibration unit's default calibration 
 kit file and ports 2, 4, and 1 of the cal unit. </p>
<p class=Example><span class=dropspot>CORR:COLL:AUTO:PORT:CONN? </span></p>
<p class=Exampletext>Query the actual port assignment. If the cal unit 
 is properly connected according to the previous command, the response 
 is <span style="font-style: italic;"><I>1,2,2,4,3,0,4,1.</I></span> A zero means 
 that the corresponding analyzer port is not connected to any port of the 
 calibration unit. </p>
<p class=space>&nbsp;</td></tr>
</table>


<hr align="center" width="50%" class="whs3">


<h6><a name=CONNection></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:CONNection&lt;port_no&gt;</span><br>
N50FEMALE | N50MALE | N75FEMALE | N75MALE | PC7 | PC35FEMALE | PC35MALE 
 | PC292FEMALE | PC292MALE</h6>

<h6>Additional parameters (<img src="../../small_images/ZVR.gif" x-maintain-ratio="TRUE" width="47px" height="16px" border="0" class="img_whs4">): UFEMALE1 | UMALE1 | UFEMALE2 
 | UMALE2 | SMAFEMALE | SMAMALE </h6>

<p>Selects a connector type at a specified port &lt;port_no&gt; and its 
 gender. </p>

<p><img src="../../small_images/tip.jpg" x-maintain-ratio="TRUE" width="36px" height="37px" border="0" class="img_whs5"> Use <span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#SCONnection"
	id=A51
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:SCONnection&lt;port_no&gt;</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A51');
//--></script><span 
 class=dropspot> to select an arbitrary connector type using a string variable. 
 </span></p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;port_no&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Port numbers of the analyzer. </p>
<p class=Note><span style="font-weight: bold;"><B>Note: </B></span>If the analyzer 
 is set to use the same connectors at all ports (<span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#CONNection_PORTs"
	id=a52
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a52');
//--></script></span><a 
 href="SENSe_CORRection_COLLect.htm#CONNection_PORTs"
	id=A7
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ "><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:CONNection&lt;port_no&gt;:PORTs 
 ALL</span></a><script type="text/javascript"
											language=JavaScript1.2><!--
kadovInitEffects('A7');
//--></script><span 
 class=dropspot>), then a change of a connector type is valid for all ports. 
 The gender of the connectors can still be different.</span> </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Parameters</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Connector type and gender of the connectors (omitted for 
 query). UFEMALE1 and UMALE1 denote the user-defined connector type <span 
 style="font-style: italic;"><I>UserConn1, </I></span>UFEMALE2 and UMALE2 denote 
 the user-defined connector type <span style="font-style: italic;"><I>UserConn2.</I></span> 
 SMAFEMALE and SMAMALE denote the user-defined connector type <span style="font-style: italic;"><I>SMA.</I></span> 
 The user-defined connector types must be defined before being addressed 
 by <span style="font-family: 'Courier New', monospace;">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:CONNection&lt;port_no&gt;.</span></td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>N50FEMALE for all ports. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, command or query.</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs8">
<p class=Example><span class=dropspot>*RST; CORR:COLL:CONN1 N75MALE; CONN4? 
 </span></p>
<p class=Exampletext>Change the connector type at port 1 from <span style="font-family: 'Courier New', monospace;">N50FEMALE</span> 
 to <span style="font-family: 'Courier New', monospace;">N75MALE</span>. 
 The connector type at the other ports is also changed to N75, however, 
 the gender (female) is maintained. <span class=dropspot
											style="font-family: 'Courier New', monospace;">CORR:COLL:CONN4?</span> 
 returns <span style="font-family: 'Courier New', monospace;">N75FEMALE</span>. 
 </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=CONNection_PORTs></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:CONNection:PORTs<br>
ALL | SINGle</span></h6>

<p>Qualifies whether the connector types at the analyzer ports (but not 
 their gender) are equal or independent. </p>

<p class=Note><img src="../../small_images/note.jpg" x-maintain-ratio="TRUE" width="36px" height="37px" border="0" class="img_whs5">In the present firmware version, a <span style="font-weight: bold;"><B>calibration</B></span> 
 must be performed using the same connector type at all ports. Use [SENSe&lt;Ch&gt;:]<span 
 class=dropspot>CORRection:COLLect:CONNection&lt;port_no&gt;:PORTs SINGle 
 if you want to perform a </span><span class=dropspot
										style="font-weight: bold;"><B>measurement</B></span><span 
 class=dropspot> with independent port connectors.</span> </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Parameters</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>ALL: Equal (uniform) connector types. If the connector 
 type at one port is changed, the connector type at all other ports is 
 changed accordingly. <br>
SINGle: Independent (possibly non-uniform) connector types at the ports. 
 &nbsp;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;port_no&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Port numbers of the analyzer. This parameter has no effect 
 because the setting affects all ports. &nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>ALL </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, command or query.</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs9">
<p class=Example>CORR:COLL:CONN:PORTS SING</p>
<p class=Exampletext>Select independent connector types at the ports.</p>
<p class=Example><span class=dropspot>CORR:COLL:CONN1 N50MALE; CONN4 N75FEMALE; 
 CONN2? </span></p>
<p class=Exampletext>Select independent connector types at ports 1 and 
 4. The connector type at port 2 is not changed; the query returns N50FEMALE. 
 </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=DELete></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:DELete 
 ['</span>&lt;cal_name&gt;']</h6>

<p><span class=dropspot>Delet</span>es system error correction data generated 
 and stored previously<span style="font-family: 'Courier New', monospace;">. 
 </span></p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft><span class=dropspot>'</span>&lt;cal_name&gt;'</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Name of the calibration (string parameter) defined together 
 with calibration type <span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#METHod_DEFine"
	id=A17
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:METHod:DEFine</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A17');
//--></script>. 
 If nothing is specified the analyzer deletes the last system error correction 
 stored by means of <span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#SAVE"
	id=A18
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:SAVE</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A18');
//--></script><span 
 class=dropspot>. </span></td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, no query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs10">
<p class=Example>CORR:COLL:METH:DEF 'Test',RSHort,1</p>
<p class=Exampletext>Select a one-port normalization at port 1 with a short 
 standard as calibration type.</p>
<p class=Example>CORR:COLL:SEL SHOR,1</p>
<p class=Exampletext>Measure a short standard connected to port 1 and store 
 the measurement results of this standard. </p>
<p class=Example>CORR:COLL:SAVE; DEL </p>
<p class=Exampletext>Calculate the system error correction data and apply 
 them to the active channel, then delete the data. </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=METHod></a><img src="../../small_images/ZVR.gif" x-maintain-ratio="TRUE" width="47px" height="16px" border="0" class="img_whs4"> [SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:METHod 
 </span><br>
FRTRans | TOM | TRM | TRL | TNA | TOSM | FOPort1 
 | FOPort2 | FOPort12 | FOPTport | ROPTport | REFL1 | REFL2 | REFL12 | 
 TPORt | UOSM</h6>

<p>Selects a one-port or two-port calibration type at ports 1/2. </p>

<p><img src="../../small_images/tip.jpg" x-maintain-ratio="TRUE" width="36px" height="37px" border="0" class="img_whs5"> Use the generalized command <span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#METHod_DEFine"
	id=A12
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:METHod:DEFine</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A12');
//--></script> 
 to select the calibration type for arbitrary analyzer ports or a multiport 
 calibration type. </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Parameters</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Calibration types: <span style="font-style: italic;"><I>TOM, 
 TRM, TRL, TNA, TOSM, Full One Port, One Path Two Port, Normalization </I></span>(REFL1, 
 REFL2 and REFL12 for one-port, TPORT for two-port)<span style="font-style: italic;"><I>, 
 </I></span><span style="font-style: italic;"><I>TOSM 
 with unknown through</I></span>. &nbsp;</p>
<p class=Tabtext>The numbers in the parameters denote the analyzer ports. 
 Parameters for two-port calibration types contain no numbers because the 
 command is only valid for ports 1 and 2.</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Confirmed (with device-specific calibration types), command 
 or query. If several calibration types are assigned to channel &lt;Ch&gt;, 
 the query returns a list of all calibration types. &nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs6">
<p class=Tabtext><span style="font-family: Arial, sans-serif;">See</span><span 
 style="font-family: 'Courier New', monospace;"> </span><a href="#SAVE"
															id=a28
															style="color: #0000ff;
	font-family: 'Courier New', monospace;
	/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]<span 
 class=dropspot
	style="color: #0000ff; font-family: 'Courier New', monospace;"><font color=blue >CORRection:COLLect:SAVE</font></span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a28');
//--></script><span 
 style="font-family: 'Courier New', monospace;">. </span></p>
<p class=space>&nbsp;</td></tr>
</table>


<hr align="center" width="50%" class="whs3">


<h6><a name=METHod_DEFine></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:METHod:DEFine<br>
'</span>&lt;cal_name&gt;', REFL | RSHort | FOPort | FRTRans | OPTPort | 
 TOSM | TOM | TRM | TRL | TNA | UOSM, &lt;port_no&gt;[,&lt;port_no&gt;][,&lt;port_no&gt;][,&lt;port_no&gt;]</h6>

<p>Selects a one-port, two-port, three-port or four-port calibration type 
 at arbitrary analyzer ports. </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>'&lt;cal_name&gt;'</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Name of the calibration (string parameter). The name serves 
 as a reference to delete a particular set of system correction data (<span 
 style="font-family: 'Courier New', monospace;"><a href="SENSe_CORRection_COLLect.htm#DELete"
													id=A14
													style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:DELete</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A14');
//--></script>). 
 </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Parameters</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Calibration types: <span style="font-style: italic;"><I>One-port 
 Normalization (reflection), using an open </I></span>(REFL)<span style="font-style: italic;"><I> 
 or a short (RSHort) standard, Full One Port, Two-port Normalization (transmission,</I></span> 
 FRTRans)<span style="font-style: italic;"><I>, One Path Two Port, TOSM</I></span>,<span 
 style="font-style: italic;"><I> TOM, TRM, TRL, TNA, </I></span><span style="font-style: italic;"><I>TOSM 
 with unknown through</I></span>. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;port_no&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Port numbers of the analyzer. For an n-port calibration 
 type (n = 1 to 4), n port numbers must be specified. If more than n numbers 
 are defined, the spare numbers (the last ones in the list) are ignored. 
 Entering less than n numbers causes an error message. </td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, no query. Use <span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#METHod"
	id=A46
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:METHod?</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A46');
//--></script><span 
 class=dropspot> t</span>o obtain a list of all calibration types for channel 
 &lt;Ch&gt;. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs7">
<p class=Example
	style="font-family: 'Courier New', monospace;"><span style="font-family: Arial, sans-serif;">See</span> 
 <span><a href="#SAVE_SELected"
			id=a16
			style="color: #0000ff; /*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="color: #0000ff;"><font color=blue >CORRection:COLLect:SAVE:SELected</font></span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a16');
//--></script>. 
 </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=SAVE></a><img src="../../small_images/ZVR.gif" x-maintain-ratio="TRUE" width="47px" height="16px" border="0" class="img_whs4"> [SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:SAVE</span></h6>

<p>Calculates the system error correction data from the acquired one or 
 two-port measurement results (<span style="font-family: 'Courier New', monospace;"><a 
 href="#AQUire"
	id=A8
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect[:ACQuire]</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A8');
//--></script>),</span><span 
 class=dropspot> stores them and applies them to the calibrated channel 
 &lt;Ch&gt;</span>. To avoid incompatibilities, older system error correction 
 data is deleted unless it has been transferred into a cal pool (<span 
 style="font-family: 'Courier New', monospace;"><a href="../MMEMory/MMEMory.htm#STORe_CORRection"
													id=A10
													style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">MMEMory:STORe:CORRection 
 &lt;Ch&gt;, '&lt;file_name&gt;'</a><script type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A10');
//--></script>)</span><span 
 style="font-family: Arial, sans-serif;">.</span></p>

<p class="whs11"><img src="../../small_images/tip.jpg" x-maintain-ratio="TRUE" width="36px" height="37px" border="0" class="img_whs5"><span style="font-family: Verdana, sans-serif;"> 
 This command is the ZVR-compatible equivalent of</span> <span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#SAVE_SELected"
	id=A25
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:SAVE:SELected</span></a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A25');
//--></script><span 
 class=dropspot>.</span> <span style="font-family: Verdana, sans-serif;">It 
 must be used in combination with the ZVR-compatible commands for calibration 
 method and standard selection; see example below.</span></p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Confirmed, no query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs10">
<p class=Example>CORR:COLL:METH REFL1</p>
<p class=Exampletext>Select a one-port normalization at port 1 as calibration 
 type.</p>
<p class=Example>CORR:COLL OPEN1 </p>
<p class=Exampletext>Measure an open standard connected to port 1 and store 
 the measurement results of this standard. </p>
<p class=Example>CORR:COLL:SAVE </p>
<p class=Exampletext>Calculate the system error correction data and apply 
 them to the active channel.</p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=SAVE_DEFault></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:SAVE:DEFault</span></h6>

<p><span class=dropspot>Generate</span>s a set of default system error 
 correction data for the selected ports and calibration type. The default 
 data set corresponds to a test setup which does not introduce any systematic 
 errors; none of the measurement results acquired previously (<span style="font-family: 'Courier New', monospace;"><a 
 href="#AQUire"
	id=a31
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect[:ACQuire]</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a31');
//--></script>)</span> 
 is taken into account. </p>

<p><img src="../../small_images/tip.jpg" x-maintain-ratio="TRUE" width="36px" height="37px" border="0" class="img_whs5"> The main purpose of the default correction data set is 
 to provide a dummy system error correction which you can replace with 
 your own, external correction data. You may have acquired the external 
 data in a previous session or even on an other instrument. If you want 
 to use the external correction data on the analyzer, simply generate the 
 default data set corresponding to your port configuration and calibration 
 type and overwrite the default data. For details refer to the program 
 example below. </p>

<p><img src="../../small_images/note.jpg" x-maintain-ratio="TRUE" width="36px" height="37px" border="0" class="img_whs5"> This command must be used in combination with the ZVR-compatible 
 commands <a href="SENSe_CORRection_COLLect.htm#METHod"
			id=A32
			style="font-family: 'Courier New', monospace; /*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]<span 
 class=dropspot>CORRection:COLLect:METHod</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A32');
//--></script> 
 and </span><a HREF="SENSe_CORRection_DATA.htm#DATA"
				id=A33
				style="font-family: 'Courier New', monospace; /*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]<span 
 class=dropspot>CORRection:DATa</a><script type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A33');
//--></script>. 
 Use </span><a href="SENSe_CORRection_COLLect.htm#SAVE_SELected_DEFault"
				id=A34
				style="font-family: 'Courier New', monospace; /*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]<span 
 class=dropspot>CORRection:COLLect:SAVE:SELected:DEFault</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A34');
//--></script> 
 if you want to use </span><span 
 style="x-condition: ZVA_only;">ZVA</span><span class=dropspot>-specific 
 calibration commands or if you want to calibrate more than 2 ports. </span></p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, no query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs10">
<p class=Example>CORR:COLL:METH REFL1</p>
<p class=Exampletext>Select a one-port normalization at port 1 with an 
 open standard as calibration type.</p>
<p class=Example>CORR:COLL:SAVE:DEF </p>
<p class=Exampletext>Calculate a dummy system error correction for the 
 normalization at port 1. The dummy system error correction provides the 
 reflection tracking error term 'SCORR3'. </p>
<p class=Example>CORR:DATA? 'SCORR3' </p>
<p class=Exampletext>Query the dummy system error correction term. The 
 response is a 1 (written as 1,0 for the real and imaginary part) for each 
 sweep point (no attenuation and no phase shift between the analyzer and 
 the calibration plane). </p>
<p class=Example>CORR:DATA 'SCORR3',&lt;ASCII_data&gt;</p>
<p class=Exampletext>Replace the dummy system error correction term with 
 your own correction data, transferred in ASCII format. </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=SAVE_SELected></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:SAVE:SELected</span></h6>

<p><span class=dropspot>C</span>alculates the system error correction data 
 from the acquired measurement results (<span style="font-family: 'Courier New', monospace;"><a 
 href="#AQUire_SELected"
	id=a26
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect[:ACQuire]:SELected</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a26');
//--></script>),</span><span 
 class=dropspot> stores them and applies them to the calibrated channel 
 &lt;Ch&gt;</span>. To avoid incompatibilities, older system error correction 
 data is deleted unless it has been transferred into a cal pool (<span 
 style="font-family: 'Courier New', monospace;"><a href="../MMEMory/MMEMory.htm#STORe_CORRection"
													id=a27
													style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">MMEMory:STORe:CORRection 
 &lt;Ch&gt;, '&lt;file_name&gt;'</a><script type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a27');
//--></script>).</span></p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Confirmed, no query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs10">
<p class=Example>CORR:COLL:METH:DEF 'Test',RSHort,1</p>
<p class=Exampletext>Select a one-port normalization at port 1 with a short 
 standard as calibration type.</p>
<p class=Example>CORR:COLL:SEL SHOR,1</p>
<p class=Exampletext>Measure a short standard connected to port 1 and store 
 the measurement results of this standard. </p>
<p class=Example>CORR:COLL:SAVE:SEL </p>
<p class=Exampletext>Calculate the system error correction data and apply 
 them to the active channel.</p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=SAVE_SELected_DEFault></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:SAVE:SELected:DEFault</span></h6>

<p><span class=dropspot>Generate</span>s a set of default system error 
 correction data for the selected ports and calibration type. The default 
 data set corresponds to a test setup which does not introduce any systematic 
 errors; none of the measurement results acquired previously (<span style="font-family: 'Courier New', monospace;"><a 
 href="#AQUire_SELected"
	id=a30
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect[:ACQuire]:SELected</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a30');
//--></script>)</span> 
 is taken into account. </p>

<p><img src="../../small_images/tip.jpg" x-maintain-ratio="TRUE" width="36px" height="37px" border="0" class="img_whs5"> The main purpose of the default correction data set is 
 to provide a dummy system error correction which you can replace with 
 your own, external correction data. You may have acquired the external 
 data in a previous session or even on an other instrument. If you want 
 to use the external correction data on the analyzer, simply generate the 
 default data set corresponding to your port configuration and calibration 
 type and overwrite the default data. For details refer to the program 
 example below. </p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>&#8211;</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, no query</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs10">
<p class=Example>CORR:COLL:METH:DEF 'Test',RSHort,1</p>
<p class=Exampletext>Select a one-port normalization at port 1 with a short 
 standard as calibration type.</p>
<p class=Example>CORR:COLL:SAVE:SEL:DEF </p>
<p class=Exampletext>Calculate a dummy system error correction for the 
 normalization at port 1. The dummy system error correction provides the 
 reflection tracking error term 'REFLTRACK'. </p>
<p class=Example>CORR:CDAT? 'REFLTRACK',1,0 </p>
<p class=Exampletext>Query the dummy system error correction term. The 
 response is a 1 (written as 1,0 for the real and imaginary part) for each 
 sweep point (no attenuation and no phase shift between the analyzer and 
 the calibration plane). </p>
<p class=Example>CORR:CDAT 'REFLTRACK',1,0,&lt;ASCII_data&gt;</p>
<p class=Exampletext>Replace the dummy system error correction term with 
 your own correction data, transferred in ASCII format. </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h6><a name=SCONnection></a>[SENSe&lt;Ch&gt;:]<span class=dropspot>CORRection:COLLect:SCONnection&lt;port_no&gt;</span><br>
'&lt;conn_name&gt;', MALE | FEMale </h6>

<p>Selects a connector type at a specified port &lt;port_no&gt; and its 
 gender. In contrast to <span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#CONNection"
	id=A50
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</span><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:CONnection&lt;port_no&gt;</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('A50');
//--></script>,</span><span 
 class=dropspot> this command uses a string variable to identify the connector 
 type. </span></p>

<table x-use-null-cells>
<col>
<col>

<tr>
<td class=TableHeading>
<p class=Tableft>&lt;Ch&gt;</td>
<td class=TableText>
<p class=Tabtext>Channel number of the calibrated channel. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>&lt;port_no&gt;</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Port numbers of the analyzer. </p>
<p class=Note><span style="font-weight: bold;"><B>Note: </B></span>If the analyzer 
 is set to use the same connectors at all ports (<span style="font-family: 'Courier New', monospace;"><a 
 href="SENSe_CORRection_COLLect.htm#CONNection_PORTs"
	id=a43
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ ">[SENSe&lt;Ch&gt;:]</a><script 
 type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a43');
//--></script></span><a 
 href="SENSe_CORRection_COLLect.htm#CONNection_PORTs"
	id=a49
	style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ "><span 
 class=dropspot
	style="font-family: 'Courier New', monospace;">CORRection:COLLect:CONNection&lt;port_no&gt;:PORTs 
 ALL</span></a><script type="text/javascript"
											language=JavaScript1.2><!--
kadovInitEffects('a49');
//--></script><span 
 class=dropspot>), then a change of a connector type is valid for all ports. 
 The gender of the connectors can still be different.</span> </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Parameters</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Connector type (string parameter) and gender of the connectors. 
 The gender designation MALE | FEMale is not necessary (and ignored) for 
 sexless connector types. </td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tabright>*RST value</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>'N 50 Ohm',FEM for all ports. </p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>SCPI, Command Types</td>
<td colspan=1
	rowspan=1>
<p class=Tabtext>Device-specific, command or query.</p>
<p class=space>&nbsp;</td></tr>

<tr>
<td colspan=1
	rowspan=1>
<p class=Tableft>Example:</td>
<td colspan="1" rowspan="1" bgcolor="#CBCBCB" class="whs8">
<p class=Example><span class=dropspot>*RST; CORR:COLL:SCON1 'N 75 Ohm', 
 MALE; SCON4? </span></p>
<p class=Exampletext>Change the connector type at port 1 from '<span style="font-family: 'Courier New', monospace;">N 
 50 Ohm', FEM</span> to '<span style="font-family: 'Courier New', monospace;">N 
 75 Ohm', MALE</span>. The connector type at the other ports is also changed 
 to N 75 Ohm, however, the gender (female) is maintained. <span class=dropspot
	style="font-family: 'Courier New', monospace;">CORR:COLL:SCON4?</span> 
 returns <span style="font-family: 'Courier New', monospace;">'N 75 Ohm', 
 FEM</span>. </p>
<p class=space>&nbsp;</td></tr>
</table>

<p class=space>&nbsp;</p>


<hr align="center" width="50%" class="whs3">


<h5 class="whs12"><a href="SENSe_CORRection_COLLect.htm"
								id=a45
								style="/*begin!kadov{{*/ position: relative; /*}}end!kadov*/ "><input 
 type=image
	name=image
	src="../../small_images/baby_up.gif"
	x-save-method=compute-relative
	style="width: 20px;
			height: 23px;
			x-condition: OLH;"></a><script type="text/javascript"
	language=JavaScript1.2><!--
kadovInitEffects('a45');
//--></script></h5>

<script language="javascript1.2" type="text/javascript"><!--
kadovTextPopupInit('a15','POPUP170545336');
//--></script><script type="text/javascript" language="JavaScript" title="WebHelpRelatedTopics"><!--
if (parseInt(navigator.appVersion) >= 3 && typeof (WritePopupMenuLayer) == "function")
  WritePopupMenuLayer();
//--></script><script type="text/javascript" language="javascript1.2">
<!--
if (window.writeIntopicBar)
	writeIntopicBar(0);
//-->
</script>
</body>
</html>
